Can I adjust amplitude of Moku:Labs output reference clock ?
If you still have questions or prefer to get help directly from an agent, please submit a request.
We’ll get back to you as soon as possible.
Moku:Go Arbitrary Waveform Generator Moku:Go Data Logger Moku:Go Frequency Response Analyzer Moku:Go Logic Analyzer & Pattern Generator Moku:Go Oscilloscope & Voltmeter Moku:Go PID Controller Moku:Go Spectrum Analyzer Moku:Go Waveform Generator Moku:Go Power Supplies Moku:Go Digital Filter Box Moku:Go FIR Filter Builder Moku:Go Lock-in Amplifier
Moku:Labs output reference clock at 10MHz
Moku:Lab provides a reference clock output on the rear panel. This can be used to synchronize to other items of lab test and measurement equipment.
This clock output is fixed at 10 MHz, -3 dBm (50ohms) or 500 mVpp.
However, many of our instruments (Oscilloscope, Phasemeter, Datalogger, Phasemeter, Spectrum Analyzer) provide an in-built waveform generator. This in-built waveform generator can be used to provide a reference to external equipment; the amplitude and frequency can be adjusted.
Finally, if you are using a Moku:Lab instrument that does NOT integrate a waveform generator; you can use a second Moku:Lab to provide an adjustable amplitude reference output using our Waveform generator instrument. This second Moku:Lab can then be synchornized to the first Moku:Lab via the rear panel reference clocks.