What is the main difference between the Moku Oscilloscope and the Data Logger instruments?
If you still have questions or prefer to get help directly from an agent, please submit a request.
We’ll get back to you as soon as possible.
Moku:Go Arbitrary Waveform Generator Moku:Go Data Logger Moku:Go Frequency Response Analyzer Moku:Go Logic Analyzer & Pattern Generator Moku:Go Oscilloscope & Voltmeter Moku:Go PID Controller Moku:Go Spectrum Analyzer Moku:Go Waveform Generator Moku:Go Power Supplies Moku:Go Digital Filter Box Moku:Go FIR Filter Builder Moku:Go Lock-in Amplifier
- Python API examples
Moku's Oscilloscope is primarily used to capture snapshots of fast signals and transient waveforms. The instrument captures and displays a segment of the data once it's triggered. You can capture very fast features, but the data traces between snapshots are not continuous. The maximum sampling rate for Moku:Go's Oscilloscope is 125 MSa/s, Moku:Lab's is 500 MSa/s, and Moku:Pro's is 5 GSa/s for one channel and 1.25 GSa/s for two or more channels.
Moku's Data Logger is primarily used to capture and log continuous data. There is no gap between a single logging session and the logging duration and sampling rate can also be configured. The maximum sampling rate is capped for Moku:Go's Data Logger at 1 MSa/s for one channel and 500 kSa/s for two channels. Moku:Lab's is 1 MSa/s for one channel and 500 kSa/s for two channels when logging to the RAM, and 100 kSa/s when logging to the SD card. Moku:Pro's is 10 MSa/s for one channel, 5 MSa/s for two channels, and 2.5 MSa/s for three or more channels.