Can the Arbitrary Waveform Generator be triggered?
If you still have questions or prefer to get help directly from an agent, please submit a request.
We’ll get back to you as soon as possible.
Using the trigger on the AWG
Moku:Lab's Arbitrary Waveform Generator can be configured to trigger the generation of the output waveform.
Set the modulation type to "Burst", then it is possible to configure the trigger source. Either input 1, input 2 or the external trigger on the rear of Moku:Lab can then be selected as the trigger source.
The trigger mode can be either "Start" or "N cycle".