How to remove drift in Phasemeter measurement
If you still have questions or prefer to get help directly from an agent, please submit a request.
We’ll get back to you as soon as possible.
Moku:Go Arbitrary Waveform Generator Moku:Go Data Logger Moku:Go Frequency Response Analyzer Moku:Go Logic Analyzer & Pattern Generator Moku:Go Oscilloscope & Voltmeter Moku:Go PID Controller Moku:Go Spectrum Analyzer Moku:Go Waveform Generator Moku:Go Power Supplies Moku:Go Digital Filter Box Moku:Go FIR Filter Builder Moku:Go Lock-in Amplifier
Moku:Lab uses a digitally implemented phase-locked loop architecture to measure the phase, frequency, and amplitude of a signal.
If you are seeing an unexpected and constant drift in your phase measurement, it can be due to frequency settings at the output or the input side. You can adjust Moku:Lab's Phasemeter with the following settings to remove drift.
Output frequency setting:
First disengage "Lock" on output frequency then tap "Reacquire" to restart the measurement.
Input frequency setting:
Manually set the frequency at the Input side then tap "Reacquire" to restart the measurement.