What's the difference between external and external (PLL) mode?
If you still have questions or prefer to get help directly from an agent, please submit a request.
We’ll get back to you as soon as possible.
Moku:Go Arbitrary Waveform Generator Moku:Go Data Logger Moku:Go Frequency Response Analyzer Moku:Go Logic Analyzer & Pattern Generator Moku:Go Oscilloscope & Voltmeter Moku:Go PID Controller Moku:Go Spectrum Analyzer Moku:Go Waveform Generator Moku:Go Power Supplies Moku:Go Digital Filter Box Moku:Go FIR Filter Builder Moku:Go Lock-in Amplifier
The external mode directly multiplies the signal from input 1 and input 2. This feature is useful when the modulation signal is not a sinusoidal waveform. For example, if the modulation signal is a low duty cycle pulse, the direct multiplication significantly improves the demodulation spectral coverage, compared to sinusoidal demodulations.
The external (PLL) establishes a phase-locked loop with input 2 and generates a pair of sinusoidal waves at the same frequency for dual-phase demodulation. This allows the user to access the X, Y, R, and theta directly from the lock-in amplifier. The user can also fine-tune the phase of the local oscillator with this mode. This is the recommended external reference mode for most use cases.