External reference clock
How do I know when the Moku:Pro is using the external reference clock
If you still have questions or prefer to get help directly from an agent, please submit a request.
We’ll get back to you as soon as possible.
Moku:Go Arbitrary Waveform Generator Moku:Go Data Logger Moku:Go Frequency Response Analyzer Moku:Go Logic Analyzer & Pattern Generator Moku:Go Oscilloscope & Voltmeter Moku:Go PID Controller Moku:Go Spectrum Analyzer Moku:Go Waveform Generator Moku:Go Power Supplies Moku:Go Digital Filter Box Moku:Go FIR Filter Builder Moku:Go Lock-in Amplifier
Moku:Pro has the ability to supply an external 10 MHz reference clock to other test and measurement device. It also has an external reference clock input, which allows the Moku:Pro to lock to an external 10 MHz reference provided by another Moku or other test and measurement equipment. The reference clock input and output are on the rear panel.
In the Moku app, ensure 'alway use internal reference' is disabled. Then, when an external 10 MHz signal is applied to the Moku:Pro reference input and the Moku has locked to it, the third LED on the front panel will be blue.
More information on the front panel status LEDs can be found in the Quick Start Guide